Principle and Gas flow diagram
The sample is placed in a graphite crucible and the crucible is maintained between the upper and the lower electrode of the impulse furnace. A high current power passes through the crucible to create a high temperature. The oxide in the sample reacts with the graphite crucible, and is extracted as carbon monoxide (CO), and carried together with the carrier gas. The extracted gas is directly analysed after the dust filter. According to the oxygen concentration, it is determined by CO or CO2 after pass throw oxidizer (Copper Oxide) with a non dispersive Infra-Red analyser (NDIR).
Wide range and high accuracy measurement
*Range: Oxygen: 0-5%, Nitrogen: 0-3%, Hydrogen: 0-0.25%
*Accuracy: σn-1≦0.3ppm or CV≦1.0% for STD sample
σn-1≦0.02ppm or CV≦0.5% for Reference gas
*Accuracy H: σn-1≦0.04ppm or RSD≦2.0% for Reference gas